This manual contains installation and operation procedures for the following Page 3 . Considerations for Safe Operation of Your Biodex Multi-Joint System. Records 16 – 44 This manual covers operation procedures for the following Biodex .. Biodex Multi-Joint System 3, Rev 2 or higher, with the proper hardware. Bi doex. User Guide the Biodex for the next instruction which will be to press start the Biodex for the next instruction The PC will load the System 3 y software.
|Published (Last):||26 December 2010|
|PDF File Size:||11.76 Mb|
|ePub File Size:||14.91 Mb|
|Price:||Free* [*Free Regsitration Required]|
NEW Upper extremity attachments for hemiparetic patients. Reactive Eccentric Mode Clinical Applications1.
The Biodex System Controller front panel left and rear of unit right. The Reactive Eccentric mode may be used to perform submaximal or maximal eccentrics. The passive mode may be used for active-assisted exercise i. In this case, remove the patient immediately from the attachment by releasing the velcro cuff. A warm-up is designed to prepare an athlete. Use with Quick-Set configuration. Align shaft dot with R L. These exercises reduce stiffness and will prevent or at.
System 4 Quick-Set™
Biofeedback is provided by the high resolution color graphics monitor to encourage patient com-pliance with exercise protocols. The Isometric mode may be used near a painful range for strength carryover into the painful range. Two Biodec Cups are provided. The system will not allow selection of mode until ROM end points are set.
Allow handgrip to rotate as patient goes through motion. Retain this Owner s Manual for future reference. Service should be provided by qualified personnel only. Excellent after-sale support, installation and continuing education programs. Higher speeds in passive mode are ideal biodec active assistive exercise.
Call Biodex Customer Service first, they ll be glad to help. Dynamometer Height Lever 4. Inform patient that the test or exercise session is ready to begin. What They Are Shstem.
To remove or insert either Heel Cup, depress the Heel Cup Release Buttons and slide mnaual cupinto or out of the footplate attachment from the toe end. Remove the attachment from the dynamometer if you have not already done so. Slide dynamometer in front of ankle to be tested. In the ON position, power to the dynamometer is enabled. Release the foot pedal to lock the dynamometer in place.
BIODEX SYSTEM 3 PRO APPLICATION/OPERATION MANUAL
In eccentric exercise, the force increases as the velocity of contraction increases up to a certain point which is in contrast to concentric exercise in which the force decreases as the speed of contraction increases.
With adhesive capsulitis, the subject may be placed biodexx the passive mode with a four second pause at end range. While the following setups are standard, it should be noted that other positioning setups are pos-sible.
Inform patient that the test or exercise session is ready to begin and that the dynamometer will now allow for rotation of the dynamometer shaft once eccentric resistance meets the specified level. Use caustion when lifting the T-base to place the pads underneath. For testing consistency, it is recommended that the patient not be allowed to view the monitor.
BIODEX SYSTEM 3 PRO APPLICATION/OPERATION MANUAL – PDF
These comments come from the clinical experience of our users as well as from published journals. The passive mode may be used to carry the limb into the flexed position.
Fully ReclinedAxis of Rotation: If the Biodex System will not be used for an extendedperiod of time, you may also want to switch the controller OFF via the Main Power switch on itsrear panel.
A warm-up is designed to prepare an athlete More information. The following general procedure is provided to help clarify use of the Setup mode. Rotate the dynamometer to the yellow color code positions when setting up to test or exercise the patient s left side.
When tipping the Cable Processor Module back, after removing the toggle arm pinuse extreme caution not to drop More information. Secure appropriate stabilization straps.
Modes of Operation 3. Submaximal eccentrics may be used to protect injured or grafted structures i. Set dynamometer tilt to 0 degrees. Instruct subject to lie on positioning chair so that hip to be tested is on top. Attach appropriate fixture to dynamometer shaft, secure with locking biodeex. That’s why so many world class facilities depend on Biodex. Syetem using panel control, you must select the appropriate attachment setting in computer control the attachment setting is automatically selected when you select the joint and exercise pattern in protocol.
If desired, keep the handgrip loose to obtain active supination or pronation.
Range of motion should be limited as warranted. Dynamometer Rotation Knob 2. To rotate the seat in either direction, turn the Seat RotationHandle toward the rear of the seat.